TXY型数显式陶瓷吸水率测定仪满足中华人民共和国国家标准GB/T3810.3-2016“建筑卫生陶瓷吸水率测定方法”也满足GB3299-82“日用陶瓷吸水率测定方法”的测定条件。 Brief description: TXY digital display ceramic water absorption tester meets the national standard of the People's Republic of China GB / T3810.3-2016 "Determination method of hygienic ceramic water absorption rate" and also GB3299-82 "Determination method of water absorption of ceramics" Measurement conditions.
型数显式陶瓷吸水率测定仪采用真空法对日用陶瓷、建筑卫生陶瓷、电瓷的吸水率进行测定的仪器。TXYtype digital display ceramic water absorption testeris an instrument for measuring the water absorption of daily-use ceramics, building sanitary ceramics, and electric porcelain by vacuum method. 建筑卫生陶瓷吸水率测定方法”也满足GB3299-82“日用陶瓷吸水率测定方法”的测定条件。 Meet the national standard of the People's Republic of China GB / T3810.3-2016 " Building Sanitary Ceramic Water Absorption Measurement Method " also meet the GB3299-82 " Daily Ceramics Water Absorption Measurement Method " measurement conditions.The instrument can also be used to determine the water absorption of bricks and cement products.
Second, the main technical parameters: 1. Vacuum degree: 0.1Mpa ± 0.005Mpa adjustable
(Use imported digital display control vacuum gauge, resolution 0.1KPa, display control accuracy 0.1KPa)
4. Water injection time: 0 ～ 99 minutes and 59 seconds continuously adjustable
5. Soaking time: 0 ～ 99 hours and 59 minutes continuously adjustable 6. Power supply: 220V 180W
Structure and composition of digital display ceramic water absorption tester: TXY type digital display ceramic water absorption tester adopts mechatronics structure, which consists of vacuum container, water circulation vacuum pump, water inlet solenoid valve, water discharge valve, water container and digital controller.The water circulation vacuum pump has the special advantages of low operating noise, easy damage and low power consumption, which makes the instrument have superior performance.